Manually mounting / removing the stylus holders from SFP1
To manually mount or remove the stylus holder from the SFP1 you need to:
- Align the front sides of the probe with the stylus (look at the "Renishaw" engraving on the side of probe or alignment marks as applicable).
- Carefully align the stylus holder and the probe, allowing the magnetic attraction to gently connect the kinematic joints.
- Gently rotate the stylus holder to ensure correct location.
- To remove, securely hold the stylus holder, gently tilt the stylus holder to break the kinematic joints.