RSP3 probe change system
The REVO probe change system is designed to allow automatic REVO probe and stylus holder changing on a CMM. The primary purpose of the system is to improve flexibility with the ability to use and store stylus holders with different styli or stylus configurations.
For optimum metrology, REVO probes and stylus holders should be changed automatically using REVO change ports (RCP TC-2 and RCP2) or a flexible change rack (FCR25). These ports are mounted on the modular rack system (MRS2).
The RCP TC-2 is a thermally controlled port for changing RSP2 V2, RSP3 and SFP1 probes.
Key features of the RCP TC-2 are:
The RCP2 is configured to change RSP2 V2 stylus holders, SFP1 styli and RSH3-6 stylus holders.
Key features of the RCP2 are:
The FCR25 is a triple port unit designed to change SH25 and RSH3-1 / 2 / 3 / 4 stylus holders. A PA25-SH adaptor is used to enable a port to change SH25 stylus holders, while a PA25-RSH3 adaptor is used for changing RSH3-1 / 2 / 3 / 4 stylus holders.
Key features of the FCR25 are:
RSP3-1 / 2 / 3 / 4
NOTE: Please observe the orientation of the probe in the RCP TC-2.
NOTE: The stylus tool is used to keep the port lid open during calibration. It must be removed for normal operation.