PHC10-3 PLUS probe head controller
PH10M-iQ PLUS must be used in conjunction with PHC10-3 PLUS, UCC S3 or UCC T3 PLUS.
PHC10-3 PLUS uses an external power supply for the head, manages all head functions and communicates via a suitable interface with the CMM computer. An additional probe interface is required for probe signal conditioning when using the PHC10-3 PLUS with either a TP7 or TP200.
Powering PHC10-3 PLUS
PHC10-3 PLUS is to be used with the provided PSU - (Emerson DP4024N3M AC power adptor) www.emerson.com. This is a 24 Vdc 49 W supply and is connected to the PHC10-3 PLUS via Ø5.5 mm dc jack plug. PHC10-3 PLUS does not does require a protective earth. However, an equipotential bonding point is provided on the rear panel for connection to the rest of the installation.
- PHC10-3 PLUS is rated to operate at +24 Vdc 1 A max input current
If an internal probe interface is fitted with the PHC10-3 PLUS it will obtain its power internally from the PHC10-3 PLUS controller.
PHC10-3 PLUS rear panel layout
9-way D-type plug for PICS output
9-way D-type connector to HCU1
25-way D-type plug RS232 communications connector to CMM computer
USB type “B” socket
PHC10-3 configuration switches
15-way D-type connector to probe head
7-pin DIN raw probe connector to probe interface or multiwire input for internal interfaces
DC power jack
Equipment bond point
Refer to the PHC10-3 PLUS installation guide (Renishaw part number H-1000-0077) for further detail.
PHC10-3 PLUS LED description
Power on when lit
PHC10-3 PLUS asserting PICS STOP when lit
Head ready for use when lit
Head indexing when lit
Head datum error when lit
Head obstruct error when lit or when non PLUS interface fitted
Head overload error when lit
TP7 detected when lit
TP200 detected when lit
TP2 / TP20 / TP6 detected when lit
Probe seated when lit
Probe damped when lit