If the equipment is used in a manner not specified by the manufacturer, the protection provided by the equipment may be impaired.
There are no user serviceable parts inside the equipment.
The UCC BI controller is only warranted and approved for use with the provided PSU.
The UCC BI is isolated from ac power by disconnection of the IEC mains connector from the supplied PSU. If any additional means of isolation is required, it must be specified and fitted by the machine manufacturer or installer of the product. The isolator / disconnection device must be sited within easy reach of the operator and comply with IEC61010 and any applicable national wiring regulations for the country of installation.
WARNING: Earthing of the UCC BI is required. It is the responsibility of the OEM or installer to apply the earth bonding scheme provided by Renishaw in this installation guide.
WARNING: The system can accelerate quickly during operation. The user must not enter the working volume of the CMM until all power is removed. When within the working volume of the CMM, the user should wear eye protection.
WARNING: UCC BI is isolated from ac power by disconnection of the IEC mains connector from the supplied PSU.
WARNING: Switching off or isolating the UCC BI may NOT prevent unexpected machine movement. The user is advised to isolate the machine from the electricity supply, compressed air or other energy sources in accordance with the machine manufacturer's instructions before entering the danger zone or performing any maintenance operations.
CAUTION: It is strongly recommended that the CMM manufacturer or retrofitter includes in their maintenance instructions a periodic test of the emergency stop and, if fitted, the associated reset switch.
Prior to installation the installer must ensure that the requirements of the machinery directive are achieved by risk assessment and use of the appropriate harmonised standards (see External Documents for reference).