The 5-pin DIN socket on the TP1(S) probe provides the connections (shown below) to the probe interface.
TP2 and TP6
Connections to the probe interface are made through the M8 mounting joint.
Connections to the probe interface are made through the autojoint.
Fitting a stylus
To fit a stylus to a Renishaw touch-trigger probe, insert the correct threaded stylus or stylus adaptor into the stylus mount and tighten the stylus securely using the S7 stylus tool provided. Tightening the stylus by any means other than the stylus tool provided (e.g. spanners, drill bits, etc.) may cause internal damage to the probe mechanism.
NOTE: All stylus joints should be clean and free from dirt or debris.