|Width||440 mm (17.3 in)|
|Depth||180 mm (7.1 in)|
|Height||44 mm (1.7 in)|
|Weight||2.1 kg (4 lb 10 oz)|
UCC S5 can either be free standing or used in a 19 inch rack system.
CAUTION: Ensure the controller is disconnected from the mains supply during installation.
The UCC S5 unit draws air from the right hand side when viewed from the front and expels air out of the left hand side. A minimum clearance gap of 10 mm is necessary between the sides of the unit and any potential obstruction.
Mounting in a 19 inch rack
The rack mounting kit (part number A-1018-0189) contains two brackets and four M5 × 6 mm screws.
Assemble the brackets to the UCC S5 as shown below:
UCC S5 to SPA3 connection
The units are linked via a CAT 5 or STP/ FTP cable. 300 mm cables are supplied with the SPA3 kits. It is not recommended to use a cable over 400 mm.
Ethernet cable link to PC
This is a standard ethernet CAT 5 cross-over cable and a 5 m cable is supplied as part of the UCC S5 kit. Lengths up to a maximum of 20 m can be used.