|440 mm (17.3 in)||180 mm (7.1 in)||44 mm (1.7 in)||2.1 kg (4 lb 10 oz)|
UCC S5 can either be free standing or used in a 19 inch rack system.
The UCC S5 unit draws air from the right hand side when viewed from the front and expels air out of the left hand side. A minimum clearance gap of 10 mm is necessary between the sides of the unit and any potential obstruction.
Mounting in a 19 inch rack
The rack mounting kit (part number A-5518-0005) contains two brackets and four M5 × 6 mm screws.
Assemble the brackets to the UCC S5 as shown below:
UCC S5 to SPA3 connection
The units are linked via a CAT 5 or STP/ FTP cable. 300 mm cables are supplied with the SPA3 kits. It is not recommended to use a cable over 400 mm.
Ethernet cable link to PC
This is a standard ethernet CAT 5 cross-over cable and a 5 m cable is supplied as part of the UCC S5 kit. Lengths up to a maximum of 20 m can be used.