WARNING: Ensure the PI 7-3 is disconnected from the mains supply during installation.
WARNING: Do not exceed the operation ambient of 50 °C around the unit. Do not install near sources of heat. Forced cooling may be required in final installation.
The PI 7-3 may be mounted in a 19 inch equipment rack or may be free standing.
To mount the PI 7-3 alone requires 2 × 1/3 blanking panel kits (A-1018-0179).
If the PI 7-3 is to be placed next to a PHC10-2 use panel kit A-1018-0173
The interconnections for the TP7 system on a PH10M motorised probe head are shown in the following illustrations.
NOTE: The first illustration refers to the external PI 7-3 unit and the second showns information regarding the integral card version of the product. The table gives a list of alternative standard Renishaw cable lengths and part numbers.
Refer to the Renishaw PH10M system installation guide (H-1000-5071) for further information.
0.4 to 0.8
0.8 to 1.6
0.1 to 0.2
NOTE: Cables marked ‘unterminated' are supplied with one end prepared but unterminated. The connector parts are supplied for fitting after routing of the cable in the installation.