Installation and connection details
|Width: 440 mm (17.3 in)||Depth: 180 mm (7.1 in)||Height: 44 mm (1.7 in)||Weight: 2.1 kg (4 lb 10 oz)|
UCC T3 -2 can either be free standing or used in a 19 inch rack system.
CAUTION: Ensure the controller is disconnected from the power supply during installation.
The CMM controller unit draws air from the right hand side when viewed from the front and expels air out of the left hand side. A minimum clearance gap of 10 mm is necessary between the sides of the unit and any potential obstruction.
Mounting in a 19 inch rack
The rack mounting kit (Renishaw part number A-5518-0005) contains two brackets and four M5 × 6 mm screws. Assemble the brackets to the CMM controller as shown below:
Earth bonding scheme
Stylus change rack (SCR) connector
The SCR200 stylus change rack is connected to the controller via a 6-pin miniature DIN socket. The pin numbers are illustrated and their functions are shown in the table.
SCR200 connector (view on rear panel)
UCC T3-2 to SPA3 connection
The units must be linked by the supplied shielded CAT 5E 300 mm cable(s), no other cables are to be used.
Ethernet cable link to PC
This is a standard ethernet CAT 5E cable, a 5 m cable is supplied as part of the CMM controller kit. Lengths up to 100 metres can be used.