Installation and connection details
|440 mm (17.3 in)||180 mm (7.1 in)||44 mm (1.7 in)||2.1 kg (4 lb 10 oz)|
UCC T3 -2 can either be free standing or used in a 19 inch rack system.
The CMM controller unit draws air from the right hand side when viewed from the front and expels air out of the left hand side. A minimum clearance gap of 10 mm is necessary between the sides of the unit and any potential obstruction.
Mounting in a 19 inch rack
The rack mounting kit (Renishaw part number A-5518-0005) contains two brackets and four M5 × 6 mm screws. Assemble the brackets to the CMM controller as shown below:
Earth bonding scheme
Stylus change rack (SCR) connector
The SCR200 stylus change rack is connected to the controller via a 6-pin miniature DIN socket. The pin numbers are illustrated and their functions are shown in the table.
SCR200 connector (view on rear panel)