The SPA2-2 is powered from the ac mains supply via a connector to IEC 60320/C14. The electrical ratings of the unit are as follows:
|100 Vac - 120 Vac and 220 Vac - 240 Vac –15%, +10%||47 Hz to 63 Hz||750 W maximum|
This equipment must be connected to a protective earth conductor via a three core mains (line) cable. The mains plug shall be inserted only into a socket outlet provided with a protective earth contact. The protective earth contact shall not be negated by the use of an extension cable without protective conductor.
|Mains input connector||IEC C13 10 amp power connector|
The installer must ensure that adequate mains over-current protection is provided for the controller equipment (UCC and SPA). See the electrical requirements sections of the appropriate installation guides. This must be in accordance with any national wiring regulations applicable in the country of installation.