Manually mounting / removing the module holder from SFP2
To manually mount or remove the module holder from the SFP2 you need to:
- Align the front of the probe with the stylus (look at the "Renishaw" engraving on the side of probe or alignment marks as applicable).
- Carefully align the module holder and the probe, allowing the magnetic attraction to gently connect the kinematic joints.
- Gently rotate the module holder to ensure correct location.
- To remove, securely hold the module holder, gently tilt the module holder to break the kinematic joint.
NOTE: Manually mounting the module holder may compromise the repeatability of the tip position so is not recommended.