Fitting the SFA to the MRS2 rail / leg
The SFAs are designed to be fitted onto the Renishaw modular rack system.
It is recommended that they are attached to the MRS2 rail using the following procedure, where it is assumed that the MRS2 system is correctly installed. For full details, please refer to the MRS2 modular rack system installation guide (Renishaw part number H-1000-5255).
2. Slot the holder into the SFA extension plate, align the holes and fasten together using a 5 mm hexagonal key and two M6 × 12 screws.
NOTE: This angle selection allows for calibration to be completed with all surface finish module types.
The ability to adjust the orientation of the artefacts means that all surface finish modules, and associated geometries, can be calibrated.To attach the assembly to the SFAH-M component, align the holes accordingly then use a 6 mm hexagonal key and two M8 × 30 screws to fasten in place.
Installing the artefact assembly onto the MRS2 rail / leg
When satisfied with the plate set-up, position two M6 D-nuts as desired within the MRS2 rail or leg. Pass the M6 × 12 screws through the universal calibration plate, align with previously placed D-Nuts and then fasten using a 5 mm hexagonal key.
Fitting to the MRS2 rail
Fitting to the MRS2 leg
Fitting the SFAH-1 / SFAH-2 to the OFA
- Align the SFAH-1 / SFAH-2 with the appropriate mounting holes of the OFA and secure each one using 2 M6 × 20 screws and the 5 mm hexagonal key (supplied).