Manually mounting / removing the SFP2 probe from REVO
To manually mount or remove the probe from the REVO head you need to:
- Align the front of the head with the probe (look at the "Renishaw" engraving on the front of probe or alignment marks as applicable).
- Carefully align the probe to the head, allowing the magnetic attraction to gently connect the kinematic joints.
- Gently rotate the probe to ensure correct location.
- To remove, securely hold the probe and gently tilt to break the kinematic joints.
WARNING: Avoid touching the probe (including counter balance arms), stylus holders and other modules when the REVO-2 head LEDs are both green and the head is ready for measurement. Green LEDs indicate that the head is engaged and the controller has a valid probe calibration.