SP600 fault diagnosis
If the SP600M probing system fails to function, please contact Renishaw who will provide a test program on disk.
This program enables the operation of the AC2 card and of each probe axis to be verified:
- Status of the 3 fuses on the AC2 card
- N.B. The OVERTRAVEL ERROR flag will always be set to “1” (error state) as no overtravel unit is connected to the SP600M probe
- PROBE PRESENT verifies that the SP600M is properly connected
- Each probe axis output can be seen
If the test program works correctly but the probe axis outputs have failed or the probe present signal is not given, check system connections and the probe cable (either by replacement or for continuity). If the fault persists please contact Renishaw, as the system must be returned for repair.
If the test program does not communicate with the AC2 card at all, please ensure that the base addresses are selected correctly on the AC2 and in the test program. The user should also check that AC2 is correctly installed in the PC expansion slot. If the base address and AC2 installation is correct, please contact Renishaw for further advice.